Baxter colleague infusion pump user manual
According to the supplemental letter, Colleague infusion pumps with serial numbers 13120001CS, 13110181CC, 13120001CT, 13110338TC and higher were manufactured with an overcurrent protection circuit in the battery harness to help protect the batteries from overcurrent damage and swelling during charging.
The pump ie 7 crack for ie 64 bit windows 7 was hot to the touch, odor was present, and clear fluid was leaking from the machine.
Only the Yuasa NP2-12 (part.
BaxterColleague infusion pumps: harness available to prevent battery swelling and overheating.BaxterColleague and Syndeo infusion pumps: following device seizures by FDA, Baxter signs consent decree.An exception to this is a case in which the battery is damaged due to excessive deep discharges and the pump is operating on battery power.FDA seizes Baxter Colleague and Syndeo infusion pumps.Damaged batteries do not impact the pumps ability to function properly while operating on AC power, provided that no other failures or alarm conditions are present.This article focuses on the product recall notifications related to events reported to PA-psrs; contenido de un manual de usuario however, a number of product recalls have been issued for the Colleague pumps for other reasons, some of which led to the.S.The roller clamp was closed on the tubing, (MSO4 infusing by nurse.
In an Urgent Device Correction letter, Baxter identified a battery undercharging condition with the Colleague series pumps.4 Baxter stated that if crack dvd player windows 7 the batteries are not charged continuously for 12 hours after a Battery Low alert or a Battery Depleted alarm occurs, the remaining operating time.
Searching the facilitys PA-psrs reports to identify cases related to the Colleague pumps.
Health Devices Alerts 2005 Aug.
Health Devices Alerts 2005 Apr.
Use the panel-lockout feature as recommended in the Colleague user manual as a standard procedure during infusions.
To obtain the insulator and instructions, contact your local Baxter representative.
Following the battery management guidelines discussed in this article and in the Colleague pump user manual.The IV tubing was examined for any leakage, but none was found.Baxter Colleague Recall Notifications, the report describing fluid leaking from the pump may be related to a Baxter recall notification of long-term exposure to fluid.Install the protective insulator at the location of the ribbon cable.Contact your facilitys Biomedical or Clinical Engineering departments or Baxter to determine if the insulator is installed in your pumps.Batteries swelling could also cause internal pump damage.2 In the letter, Baxter also provided the following information: The procedure for replacement, service life, and storage conditions of the sealed lead-acid batteries used in the pumps and indicated that for Colleague single-channel monochrome-display pumps with software.Battery test and installation instructions can be found in the Colleague pump service manual.To remedy the problem, Baxter made available an insulator to be placed in the area of the ribbon cable to prevent fluid from contacting the cable.According to Baxter, following the charging instructions listed in the Colleague pump user manual will help avoid interruption or stoppage of therapy, loss of configuration memory, and/or device failure.